We have 6 Fresenius Medical Care K manuals available for free PDF download: Troubleshooting Manual, Operator’s Manual, Technical Manual. View and Download Fresenius Medical Care K operator’s manual online. K Hemodialysis Machine Operator’s Manual. K pdf manual download. FRESENIUS K OPERATORS MANUAL KHOXJMBJNW | PDF | 45 Pages | KB | 06 Apr FRESENIUS K OPERATORS MANUAL.
|Published (Last):||24 March 2009|
|PDF File Size:||7.48 Mb|
|ePub File Size:||6.31 Mb|
|Price:||Free* [*Free Regsitration Required]|
The battery gresenius will pop forward. Section 13 – Deaeration Problems Deaeration And Loading Pressure Labels And Mechanical Condition Pre-uf Pump Filter Be sure the blood pump is set to the proper ID. Don’t have an account?
FRESENIUS MEDICAL CARE K OPERATOR’S MANUAL Pdf Download.
UF rate will be maintained even with low blood pump speeds. Test Check UF pump. Page To proceed with the selected operation, close the shunt door. If necessary, replace the drain line. Powered by Social Strata. Be sure to use aseptic technique for all bloodline connections.
Other companies seem to be able to put out new equipment in the USA. Manufacturer’s Emc Declaration Figure 57 — K machine with Single Needle equipment. Other Options There are other options on these screens that may not be specifically freaenius here. The program starts fresenuus a 45 second rinse. Don’t show me this message again. About Hemodialysis Infections or pyrogen reactions may also result from contamination of the extracorporeal circuit or inadequate procedures used to reuse dialyzers.
Allergic reactions to chemical disinfectants may occur if insufficient procedures are used to remove or maintain the residual disinfectant at acceptable levels.
Fresenius Medical Care 2008K Manuals
UF Removed is mznual to 0. On Line Clearance olc Problems Acid Clean Program As the carriage moves upward, observe the heparin as it travels from the syringe through the heparin line.
Check conductivity and approximate pH with an independent device and connect dialysate lines to dialyzer in counter-current flow. Page 84 The frfsenius chamber and transducer are connected via a pressure line that runs from the chamber to venous pressure port P located on the front of the module. Temperature Sensor Calibration Volt Hi Lo Detect Wait until blood pressure module completes resetting and retry blood pressure reading.
Surprised non of you “F-ers” ever volunteered to be the lean facilitator for your clinics since it was suppossed to be a grass roots effort. General Warnings This section contains general warnings statements regarding the use and maintenance of the Fresenius K hemodialysis machine. During Treatment The machine is in fresenlus mode during the test period. General Warnings And Cautions It is now time to connect the patient to the K via the blood tubing and begin the dialysis treatment.
You should familiarize yourself with the location and purpose of these features.
Alarm in case of water shortage. Blood Pump Rotor Inspection Page 11 These instructions are for the Fresenius K Hemodialysis machine. Modules Chapter 1—Overview Modules The modules accompanying the K are located just below the control panel.
Conventions A button refers to pressure-sensitive area located in the treatment screens that is used to set treatment parameters. Page Start key a second time and the pump segment will be released from the pump head.
New Fresenius k semi & annual pm manual – Topic
Pressing the RESET key once mutes an alarm, pressing it a second time resets the right side of the screen. Heater is off during pre-rinse. How long can the fresenius t stay in chemical dwell for storage.